The mold incubator is generally used in medical and health, biopharmaceutical, agricultural scientific research, environmental protection and other research and application fields. It is a special equipment for water analysis, BOD determination, cultivation and preservation of bacteria, strains, microorganisms, plant cultivation, and breeding experimental biological culture. . The precautions required during the maintenance of the mold incubator are as follows.
1. Plug in the power socket (the power supply should be well grounded), press the power switch, the display is on, and the display shows the actual temperature and humidity inside the incubator.
2. Adjust the height with the adjustment screw at the bottom of the mold incubator to make the cabinet stable.
3. Installation of the humidifier: insert the power plug of the humidifier into the power socket on the back of the instrument, and then connect the humidifier tube of the instrument to the humidifier, and the connection must be tightly connected. The water in the humidifier water tank must be operated correctly according to the instructions.
4. Temperature adjustment: press the temperature setting button, the digital display is the set value, rotate the temperature adjustment potentiometer to the desired temperature value, release the button, the digital display is the actual temperature in the cultivation chamber. At this time, if the actual temperature in the incubator is lower than the set temperature, the heating indicator lights, and the heater begins to heat; if the actual temperature in the incubator is higher than the set temperature, the cooling indicator lights, the refrigeration system starts to cool; if the heating indicator When both the cooling indicator and the indicator light are dimmed, the incubator is in a constant temperature muffle furnace.
5. Humidity adjustment: Press the humidity setting button, the digital display is the set value, rotate the humidity adjustment potentiometer to the required humidity value, release the button, the digital display is the actual humidity in the mold incubator. When the actual humidity in the cultivation room is smaller than the set humidity, the humidification humidifies the cultivation room and the humidification indicator lights; when the actual humidity in the cultivation room is greater than the set humidity value, the humidifier stops working at this time and the humidification indicator Extinguish.
6. Care must be taken when transporting, the angle between the transport and the horizontal plane should not be less than 45 Â°.
7. When the use temperature is low, the water accumulated in the water collecting tray at the bottom of the tank should be drained regularly.
8. If there is no need for sterilization in the box, the sterilization switch on the panel should be set to the "off" position.
9. After the temperature is set, the temperature control knob cannot be rotated back and forth multiple times, so as to avoid frequent compressor startup, causing the compressor to overload and affect the life of the compressor.
10. Two sets of fuse boxes are installed on the back of the machine, 2A? Is the cooling and heating load fuse box, and 8A? Is the control power fuse box. If the machine fails, such as temperature control failure, no heating or cooling, the power must be cut off. Check whether the fuse is intact, and then check the corresponding parts.
11. If the humidifier is faulty, please follow the warranty points on the humidifier instruction manual and repair it nearby.
12. If the humidity is not used for a long time, please pour out the water in the box.
13. When the instrument is not in use, unplug the power plug. Customers should pay attention to the above points when using the mold incubator, and do the daily maintenance of the mold incubator.
14. When the humidity sensor is in a high humidity state for a long time, condensation will form, that is, the humidity display value will remain high. If an accurate humidity display value is required, the door of the incubator box should be opened after the shutdown, and the humidity sensor is at room temperature Medium, after natural drying, you can continue to use.
15. In order to maintain the beauty of the equipment, it is not allowed to wipe the surface with acid or alkali and other corrosive materials. The box can be regularly wiped dry with a dry cloth.
MODEL BASIC WEIGHT CHARACTERISTICS APPLICATION
Airthough nonowven 16-22g/m2 soft,good hydrophilic Topsheet of sanitary napkins,Baby Diapers
20-30g/m2 fluffy,soft Dry Facial Wipes and wet wipes
28-50g/m2 Quick acqusition,low rewet Acqusition layer for baby Diapers,nap-kins
40-60g/m2 fluffy,soft absorbent core and filiter core
20-30g/m2 high resilience,fluffy,well bonde-d Lamilated with PE film as tablbecloth
1.The product is with very good uniformity
2.The product is consist of continuous filament, with good tensile strength and elongation
3.Different color to meet the customers' requirement
4.Quick water penetration
5. low wet back
Hydrophilic Nonwoven,Hydrophilic Non Woven Fabric,Hydrophilic Hot Air Through Fabric,Hot Air Through Non Woven Fabric
Zhejiang Youquan Care Products Technology Co., Ltd. , https://www.youquancare.com